Load frame testing is a critical part of product design, development, and manufacture. Well designed test protocols yield the information required to determine the strength of a chosen material, and the reliability of a particular design. Through the application of a cyclic load to an object, we are able to gain an understanding of how the object will perform in actual use. The load application may be repeated millions of times at a frequency of several times per second for fatigue testing.
What equipment do you use for testing?
We perform all of our load frame testing using the Electropuls™ E10000 Linear-Torsion test frame, which is manufactured by Instron: a world leader in the material testing industry. Designed for both static and dynamic testing, it is capable of performing to more than 100Hz with a ±10kN dynamic linear load capacity, and a ±100Nm dynamic torque capacity.
What are the equipment specifications?
|Linear Dynamic Capacity||±10kN (2250lb)|
|Linear Static Capacity||±7.1kN (1600lb)|
|Torsion Capacity||±100Nm (880in-lb)|
|Linear Stroke||60mm (2.36in)|
Why is this type of testing important?
Load frame testing involves subjecting a test sample to various loading conditions to evaluate how it will perform over time. The real world effects of loading and fatigue show up as damage in the plastic and metal components of orthopaedic implants.Results of such tests allow manufacturers to make improvements needed to meet performance specifications and to create safer, stronger, more successful parts and products.
Can tests be performed under physiological conditions?
Tests can be performed in a protein-rich environment to mimic the corrosive media found in the body. For reproducing joint kinematics, please see our hip and knee simulator wear testing service pages.
Standards We Test
Standard Test Method for Determination of Total Knee Replacement Constraint
Standard Practice for Cyclic Fatigue Testing of Metal Tibial Tray Components of Total Knee Joint Replacements
Standard Test Methods for Determination of Static and Cyclic Fatigue Strength of Ceramic Modular Femoral Heads
Standard Practice for Evaluation of Modular Connection of Proximally Fixed Femoral Hip Prosthesis
Standard Test Method for Evaluating Knee Bearing (Tibial Insert) Endurance and Deformation Under High Flexion
Implants for surgery — Total knee-joint prostheses — Part 1: Determination of endurance properties of knee tibial trays
Standard Test Method for Determining the Forces for Disassembly of Modular Acetabular Devices
Standard Practice for Fretting Corrosion Testing of Modular Implant Interfaces: Hip Femoral Head-Bore and Cone Taper Interface
Standard Test Method for Determining the Axial Disassembly Force of Taper Connections of Modular Prostheses
Determination of Endurance Properties and Performance of Stemmed Femoral Components
Determination of Resistance to Static Load of Modular Femoral Heads
Standard Test Method for Evaluating Mobile Bearing Knee Tibial Baseplate/bearing Resistance to Dynamic Disassociation
Standard Test Method for Evaluating Mobile Bearing Knee Dislocation
Standard Test Method for Static Evaluation of Anatomic Glenoid Locking Mechanism in Shear
Standard Test Method for Static, Dynamic, and Wear Assessment of Extra-Discal Single Level Spinal Constructs
Standard Test Methods for Occipital-Cervical and Occipital-Cervical-Thoracic Spinal Implant Constructs in a Vertebrectomy Model
Standard Test Method for Measuring Load Induced Subsidence of Intervertebral Body Fusion Device Under Static Axial Compression
Test Methods for Intervertebral Body Fusion Devices
Standard Specification and Test Methods for Metallic Medical Bone Screws
Standard Test Methods for Spinal Implant Constructs in a Vertebrectomy Model